This Mirror Orthodontic Reflector reflects light from the flash to fully illuminate the arch. Positioned on the patients lower anterior teeth and pushed lingually to create an angle to capture occlusal imagery.
Size: 134 mm x 70 mm x 60 mm
Glass and stainless steel mirror surfaces can be cleaned with water or glass cleaner.
Dry the mirrors with a soft microfiber cloth. Do not use paper towel as it is abrasive and can dull the mirror surface overtime.
Completely dry the mirrors before placing them in an autoclave sleeve to prevent mineral spots on the mirror. Autoclaving is the recommended method of sterilisation but mirrors may also be cold sterilised.
Autoclave pouches are recommended to keep the mirrors sterile until needed.
Autoclave recommended settings: 121°C (249.8°F) for 20+ minutes or134°C (273.2°F) for 10+ minutes.
Do not put mirrors into an ultrasonic cleaning device and do not use bleach solutions. Discard any mirror with a chipped or scratched edge.